Detectors.page 9.41 KB
 Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 1 * [S3 specs](/s-detectors-specs.pdf)  Oliver Kirsebom committed Jun 12, 2017 2   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 3 * [S3 PCB Dimensions](/A-3329.pdf)  Oliver Kirsebom committed Mar 03, 2017 4   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 5 6 * [Detailed drawing of W1 profile (with dimensions)](/w1 profile p2.pdf). According to Neil from Micron Semiconductors, the height of the oxide is $\approx 1$ $\mu$m and the depth of the deep doping is $\approx 500$ nm.  Oliver Kirsebom committed Mar 03, 2017 7   Oliver Kirsebom committed Nov 10, 2017 8 * NIM paper from 2013 discussing interstrip effects in W1 detectors: [Torresi et al.](/torresi2013.pdf)  Oliver Kirsebom committed Nov 10, 2017 9   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 10 11 12 13 14 * Neil could not provide us with a detailed drawing of the S3 profile. However, he gave this description: The S3 detector is a little more complicated in terms of the layer structure on the front when compared to the W1 as uses a double metal process. However, the junction implant is only a type 2 with a 0.5um depth. On top of silicon junction is metal 1 at 0.6 um, then 3.5 um of PECVD oxide to separate this from M2 which is 1.5 um where it occurs. So total of 6.1 um where metal 2 is (just fan-our areas) and 4.6 um on most of front surface. Back is the same as a W1 with 0.2 - 0.3 um metal on the 0.4 um implant layer. He also informed us that the inter-spoke spacing is 100 um (with a central 40 um common p-stop between spokes).  Oliver Kirsebom committed Mar 03, 2017 15   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 16 * **[Link to Micron's detailed spec overview](/micron.pdf)**.  Oliver Kirsebom committed May 02, 2016 17   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 18 # W1 (MSX25 compatible)  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 19 20 21  [Technical drawing](detector_spec/drawings/w1.pdf)  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 22 23 24 25 26 27 28 29 30 31 32 33 34 | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:-----------:|:-------------:|:------------------:|:--------------:|:-----:|:-------------------------:|:---------------------:|:---------------------------------:|:-----------------------------------:| | TBD | 3203-11 | 2016 | 40 | 9G/2M | 12 | Aarhus | [link](detector_spec/3203-11.pdf) | | | AUW1_50_00 | 3189-3 | 2015 | 55 | 9G/2M | 9 | Aarhus | [link](detector_spec/3189-3.pdf) | | | AUW1_50_01 | 3188-5 | 2015 | 55 | 9G/2M | 20 | Aarhus | [link](detector_spec/3188-5.pdf) | | | AUW1_60_00 | ? | 2008? | ~60 | ?G/2M | ? | Aarhus | - | From I. E. M. Madrid; used in IS999 | | AUW1_60_02 | 2887-4 | 2011 | 65 | 9G/2M | 10 | Aarhus | [link](detector_spec/2887-4.pdf) | | | AUW1_60_01 | 2881-1 | 2011 | 67 | 9G/2M | 10 | Aarhus | [link](detector_spec/2881-1.pdf) | | | AUW1_65_00 | 2846-5 | 2017 | 68 | 2M/2M | 10 | ? | [link](detector_spec/2846-5.pdf) | | | AUW1_65_01 | 3187-1 | 2017 | 70 | 2M/2M | 24 | ? | [link](detector_spec/3187-1.pdf) | | | AUW1_500_00 | 2495-18 | 2012 | 524 | 9G/2M | 40 | Aarhus | [link](detector_spec/2495-18.pdf) | | | TBD | 2541-7 | 2015 | 1002 | ?G/2M | 85 | Aarhus | [link](detector_spec/2541-7.pdf) | | | TBD | 2947-4 | 2017 | 1043 | ?G/2M | 146 | Aarhus | [link](detector_spec/2947-4.pdf) | |  Munken committed Sep 07, 2016 35 36  # S3  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 37   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 38 [Technical drawing 1](detector_spec/drawings/s3.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 39   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 40 [Technical drawing 2 (special connector)](detector_spec/drawings/s3_special.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 41   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 42 43 44 45 46 47 48  | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:---------------------------------:|:-------------:|:------------------:|:--------------:|:------:|:-------------------------:|:---------------------:|:---------------------------------:|:-----------------:| | AUS3_60_00 | 2915-5 | 2011 | 64 | 2DM/2M | 14 | Aarhus | [link](detector_spec/2915-5.pdf) | Regular connector | | [AUS3_322_00](dets/AUS3_322_00) | 3040-2 | 2013 | 322 | 2DM/2M | 22 | Aarhus | [link](detector_spec/3040-2.pdf) | Special connector | | [AUS3_1000_00](dets/AUS3_1000_00) | 2807-21 | 2012 | 1038 | 2DM/2M | 175 | Aarhus | [link](detector_spec/2807-21.pdf) | Regular connector | | AUS3_1000_01 | 2807-22 | 2012 | 1039 | 2DM/2M | 155 | Aarhus | [link](detector_spec/2807-22.pdf) | ? connector |  Munken committed Sep 07, 2016 49   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 50 # Pads / MSX25 (modified W1 design)  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 51   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 52 53 [Technical drawing](detector_spec/drawings/msx25_special.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 54   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 55 56 57 58 59 60 61 62 63 | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:-------------:|:-------------:|:------------------:|:--------------:|:------:|:-------------------------:|:---------------------:|:--------------------------------:|:-------------------------------------------------:| | - | 3145-8 | 2014 | 1029 | ? | 175 | ? | [link](detector_spec/3145-8.pdf) | | | - | 3145-6 | 2014 | 1042 | ? | 80 | ? | [link](detector_spec/3145-6.pdf) | | | AUPAD_1458_00 | - | ? | 1458 | 2M/2M? | ? | ? | | | | AUPAD_1500_02 | 3289-5 | 2017 | 1490 | ? | 260 | Aarhus | [link](detector_spec/3289-5.pdf) | | | AUPAD_1500_01 | 3289-4 | 2017 | 1497 | ? | 260 | Aarhus | [link](detector_spec/3289-4.pdf) | | | TBD | - | ? | 1498 | 2M/2M? | 30? | Aarhus | | From I. E. M. Madrid; duplicate of AUPAD_1458_00? | | MD0002 | 2280-27 | 2008? | 1510 | 2M/2M? | ? | Aarhus | | From I. E. M. Madrid; big scratch |  Erik Asbjørn Mikkelsen Jensen committed May 31, 2021 64   Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 65 66 # BB7  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 67 [Technical drawing](detector_spec/drawings/bb7.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 68   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 69 70 71 | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:--------------------------------:|:-------------:|:------------------:|:--------------:|:-----:|:-------------------------:|:---------------------:|:--------------------------------:|:---------------------------------:| | [AUBB7_140_00](dets/AUS3_322_00) | 3228-6 | 2016 | 140 | ??/2M | 15 | ? | [link](detector_spec/3228-6.pdf) | Maybe Oliver Kirsebom knows more? |  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 72   Erik Asbjørn Mikkelsen Jensen committed May 31, 2021 73 # Clovers at IDS  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 74   Erik Asbjørn Mikkelsen Jensen committed May 31, 2021 75 76 The Clover detectors used at the Isolde Decay Station are of the so-called EUROBALL type.  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 77 78 Each detector consists of four HPGe crystals, and each crystal has a diameter of 50 mm and a length of 70 mm (see page 3 in [this document](/c39840_clover_detector_super_spec_3.pdf)).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 79   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 80 # W1 (MSX25 incompatible)  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 81   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 82 [Technical drawing](detector_spec/drawings/w.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 83   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 84 85 86 | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:---:|:-------------:|:------------------:|:--------------:|:-----:|:-------------------------:|:---------------------:|:---------------------------------:|:--------:| | - | 3106-10 | 2016 | 170 | 9T/2M | 19 | Aarhus | [link](detector_spec/3106-10.pdf) | |  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 87 88 89  # MSX25 (regular)  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 90 91 [Technical drawing](detector_spec/drawings/msx25.pdf).  Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 92   Erik Asbjørn Mikkelsen Jensen committed Dec 12, 2021 93 94 95 | ID | WAFER
NO. | YEAR
PURCHASED | THICKNESS (um) | TYPE | DEPLETION
VOLTAGE (V) | PHYSICAL
LOCATION | SPEC
SHEET | COMMENTS | |:---:|:-------------:|:------------------:|:--------------:|:------:|:-------------------------:|:---------------------:|:---------------------------------:|:--------:| | - | 2826-7 | 2013 | 524 | 2M/2M? | 36 | Aarhus | [link](detector_spec/2826-7.pdf) | |