Detectors.page 5.94 KB
 Oliver Kirsebom committed Jun 12, 2017 1 * [S3 specs](/s-detectors-specs.pdf)  Oliver Kirsebom committed Jun 12, 2017 2   Oliver Kirsebom committed Feb 16, 2017 3 * [S3 PCB Dimensions](/A-3329.pdf)  Oliver Kirsebom committed Mar 03, 2017 4 5 6  * [Detailed drawing of W1 profile (with dimensions)](/w1 profile p2.pdf). According to Neil from Micron Semiconductors, the height of the oxide is $\approx 1$ $\mu$m and the depth of the deep doping is $\approx 500$ nm.  Oliver Kirsebom committed Nov 10, 2017 7 * NIM paper from 2013 discussing interstrip effects in W1 detectors: [Torresi et al.](/torresi2013.pdf)  Oliver Kirsebom committed Nov 10, 2017 8   Oliver Kirsebom committed Jun 13, 2017 9 * Neil could not provide us with a detailed drawing of the S3 profile. However, he gave this description: The S3 detector is a little more complicated in terms of the layer structure on the front when compared to the W1 as uses a double metal process. However, the junction implant is only a type 2 with a 0.5um depth. On top of silicon junction is metal 1 at 0.6 um, then 3.5 um of PECVD oxide to separate this from M2 which is 1.5 um where it occurs. So total of 6.1 um where metal 2 is (just fan-our areas) and 4.6 um on most of front surface. Back is the same as a W1 with 0.2 - 0.3 um metal on the 0.4 um implant layer. He also informed us that the inter-spoke spacing is 100 um (with a central 40 um common p-stop between spokes).  Oliver Kirsebom committed Mar 03, 2017 10   Andreas Gad committed Mar 03, 2021 11 * Link to Microns own spec [sheets](/micron.pdf).  Oliver Kirsebom committed May 02, 2016 12   Munken committed Sep 07, 2016 13 # W1  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 14 15 16  [Technical drawing](detector_spec/drawings/w1.pdf)  Munken committed Sep 07, 2016 17 18 19 20 21 22 23 24 | ID | YEAR PURCHASED | SPEC SHEET | THICKNESS (um) | DEADLAYER F/B (um) | BIAS (V) | LEAKAGE (uA) | JSON FILE | PHYSICAL LOCATION | COMMENTS | | :--- | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | | AUW1_50_00 | - | - | 55 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_50_00.json) | Aarhus | - | | AUW1_50_01 | - | - | 55 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_50_01.json) | Aarhus | - | | AUW1_500_00 | - | - | 524 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_500_00.json) | Aarhus | - | | AUW1_60_00 | 2008 | link | 65 | -/- | -40 | 0.300 | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_60_00.json) | Aarhus | used in IS999 | | AUW1_60_01 | | | ~60 | -/- | -20 | 0.189 | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_60_01.json) | Aarhus | | | AUW1_60_02 | | | ~60 | -/- | -20 | 0.189 | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUW1_60_02.json) | Aarhus | |  Oliver Kirsebom committed Dec 19, 2017 25 26 | AUW1_65_00 | 2017 | [link](/AUW1_65_00_specs.pdf) | 68 | -/- | - | - | - | Aarhus | Wafer no. 2846-5. Not tested yet | | AUW1_65_01 | 2017 | [link](/AUW1_65_01_specs.pdf) | 70 | -/- | - | - | - | Aarhus | Wafer no. 3187-1. Not tested yet |  Munken committed Sep 07, 2016 27 28  # S3  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 29 30  [Technical drawing 1](detector_spec/drawings/s3.pdf)  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 31   Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 32 33 [Technical drawing 2](detector_spec/drawings/s3_special.pdf)  Munken committed Sep 07, 2016 34 35 | ID | YEAR PURCHASED | SPEC SHEET | THICKNESS (um) | DEADLAYER F/B (um) | BIAS (V) | LEAKAGE (uA) | JSON FILE | PHYSICAL LOCATION | COMMENTS | | :--- | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: |  Munken committed Sep 07, 2016 36 | [AUS3_322_00](dets/AUS3_322_00) | 2005 | - | 320 | -/- | -200 | 1.000 | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUS3_322_00.json) | Aarhus | ring 7 dead |  Munken committed Sep 07, 2016 37 | [AUS3_1000_00](dets/AUS3_1000_00) | - | - | 1038 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUS3_1000_00.json) | Aarhus | - |  Jesper Halkjær Jensen committed Sep 22, 2016 38 | AUS3_1000_01 | 2016 | [link](/AUS3_1000_01.pdf) | 1039 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUS3_1000_01.json) | Aarhus | - |  Munken committed Sep 07, 2016 39 40 | AUS3_60_00 | - | - | 64 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUS3_60_00.json) | Aarhus | - |  Jesper Halkjær Jensen committed Feb 20, 2017 41 # BB7  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 42 43 44  [Technical drawing](detector_spec/drawings/bb7.pdf)  Jesper Halkjær Jensen committed Feb 20, 2017 45 46 | ID | YEAR PURCHASED | SPEC SHEET | THICKNESS (um) | DEADLAYER F/B (um) | BIAS (V) | LEAKAGE (uA) | JSON FILE | PHYSICAL LOCATION | COMMENTS | | :--- | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: |  Jesper Halkjær Jensen committed Feb 20, 2017 47 | [AUBB7_140_00](dets/AUS3_322_00) | 2017 | [link](/AUBB7_140_00.pdf) | 140 | -/- | -60 | 0.1 | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUBB7_140_00.json) | Aarhus | - |  Munken committed Sep 07, 2016 48 49  # Pads  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 50 51 52  [Technical drawing](detector_spec/drawings/msx25_special.pdf)  Munken committed Sep 07, 2016 53 54 55 | ID | YEAR PURCHASED | SPEC SHEET | THICKNESS (um) | DEADLAYER F/B (um) | BIAS (V) | LEAKAGE (uA) | JSON FILE | PHYSICAL LOCATION | COMMENTS | | :--- | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | :---: | | AUPAD_1458_00 | - | - | 1458 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/AUPAD_1458_00.json) | Aarhus | - |  Jesper Halkjær Jensen committed Nov 06, 2017 56 | MD0002 | - | - | ~1500 | -/- | - | - | [link](https://git.kern.phys.au.dk/ausa/ausalib/blob/master/res/Detectors/MD0002.json) | Aarhus | Big scratch |  Jesper Halkjær Jensen committed Nov 06, 2017 57 | AUPAD_1500_01 | 2017 | [link](/pads_data_sheets_OK_sep2017.pdf) | 1497 | -/- | - | - | - | Aarhus | |  Erik Asbjørn Mikkelsen Jensen committed May 31, 2021 58 59 60 61 62 | AUPAD_1500_02 | 2017 | [link](/pads_data_sheets_OK_sep2017.pdf) | 1490 | -/- | - | - | - | Aarhus | | # Clovers at IDS The Clover detectors used at the Isolde Decay Station are of the so-called EUROBALL type.  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 63 64 65 66 67 Each detector consists of four HPGe crystals, and each crystal has a diameter of 50 mm and a length of 70 mm (see page 3 in [this document](/c39840_clover_detector_super_spec_3.pdf)). # Others [Technical drawing of regular MSX25](detector_spec/drawings/msx25.pdf)  Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 68   Erik Asbjørn Mikkelsen Jensen committed Dec 11, 2021 69 [Technical drawing of different W design](detector_spec/drawings/w.pdf)